There is an urgent need for reliable high-throughput serological assays for the management of the ongoing COVID-19 pandemic. Preferably, the performance of serological tests for a novel virus should be determined with clinical specimens against a gold standard, i.e. virus neutralisation. We compared the performance of six commercial immunoassays for the detection of SARS-COV-2 IgG, IgA and IgM antibodies, including four automated assays [Abbott SARS-COV-2 IgG (CE marked), Diasorin Liaison® SARS-COV-2 S1/S2 IgG (research use only, RUO), and Euroimmun SARS-COV-2 IgG and IgA (CE marked)], and two rapid lateral flow (immunocromatographic) tests [Acro Biotech 2019-nCoV IgG/IgM (CE marked) and Xiamen Biotime Biotechnology SARS-COV-2 IgG/IgM (CE marked)] with a microneutralisation test (MNT). Two specimen panels from serum samples sent to Helsinki University Hospital Laboratory (HUSLAB) were compiled: the patient panel (N=70) included sera from PCR confirmed COVID-19 patients, and the negative panel (N=81) included sera sent for screening of autoimmune diseases and respiratory virus antibodies in 2018 and 2019. The MNT was carried out for all COVID-19 samples (70 serum samples, 62 individuals) and for 53 samples from the negative panel. Forty-one out of 62 COVID-19 patients showed neutralising antibodies.The specificity and sensitivity values of the commercial tests against MNT, respectively, were as follows: 95.1 %/80.5 % (Abbott Architect SARS-CoV-2 IgG), 94.9 %/43.8 % (Diasorin Liaison SARS-CoV-2 IgG; RUO), 68.3 %/87.8 % (Euroimmun SARS-CoV-2 IgA), 86.6 %/70.7 % (Euroimmun SARS-CoV-2 IgG), 74.4 %/56.1 % (Acro 2019-nCoV IgG), 69.5 %/46.3 % (Acro 2019-nCoV IgM), 97.5 %/71.9 % (Xiamen Biotime SARS-CoV-2 IgG), and 88.8 %/81.3 % (Xiamen Biotime SARS-CoV-2 IgM). This study shows variable performance values. Laboratories should carefully consider their testing process, such as a two-tier approach, in order to optimize the overall performance of SARS- CoV-2 serodiagnostics.